- Vatech 3D Viewers
- Hardware Sensors
- Hardware Capture Program
- Legacy Devices
- General Q&A
Please use the following search box to quickly locate valuable FAQ support guide.You can use key words or full statement to search by. Alternatively you can simply browse by Category below
Datagrabber & SDK integration Compatibility for Vatech Software - The following chart to be used to identify
Note: Not all programs capable of twaining are listed, those listed are simply one's that have been confirmed to work. Most twain compliant programs will support twain* generally with out issue. Please contact those software's that are not listed to confirm they support twain
|Source Software A - K||Special Notes:||To Apteryx/DG Bridge||To EasyDent4 (SDK)||To EzDent I (SDK)||NCSW (SDK/Twain)||Intra Oral SDK/Twain Auto Advance|
|ABEL Dent||F, DIA-XV|
|Ace||Users MUST contact ACE to setup bridge; we will not supply Datagrabber for ACE users. ACE has SDK integration with Easydent4 and can make it work with EZDent-i though there is a charge from their support. NOTE: Data grabber can be supplied to older ACE users -only using the MDB type not the updated SQL based*||Full Integration||Bridge||Bridge|
|Apteryx - XrayVision||(XrayVision, OneView; Prof Suni, ImageXL)||Bridge||Bridge||Bridge||Bridge/Twain||AutoAdvance|
|Bridge IT||Full Integration|
|CareStream "CS" software||Carestreamconfirmed bug on the 7.0.1 version it will not take twain.
Please advise office to downgrade to an earlier build when you see this until update is availilbe.
|CareStack||Pano images - useI/O for Sensors||Bridge||Twain||AutoAdvance|
|CIEOS||Bridge available for older versions*||Bridge||Bridge||Twain||Twain|
|CleoSoft||Need to confirm*|
|Computer Age Dentist||Full Integration|
|Cloud 9||Setup by Cloud 9 Bridge/Twain||Twain||Twain|
|Curve Dental||Direct Integration for Sensors availible* and twain for P2 units||Twain||Twain|
|Curve (Hero)||For twain they must purchase imaging contract||Twain||Twain|
|Denticom||Web based - IE - Active X SDK directly from their support*||BridgeTwain|
|DentalVision||Henry Schein | Easydent4 only.||S||Bridge|
|Dentech by Softech||DIA-XV|
|Denti Max||*Latest version requires special setup by Dentixmax; requiring the following folder and file to be inplace on all workstations (C:\VCapturesw\exe\NCSW_2012.exe) only the one .exe is required no other files* can simply copy and paste and call to dentimax for the integration to be setup. Also; for updated VCaptureSW.exe versions dentimax will need to update their link.||Full Integration, DIA-XV||Bridge||Twain||Twain|
|Denti Soft||Full Integration|
|Dentrix Enterprise & IMAGING||DataGrabber only*||Full Integration||Bridge||Twain||Twain|
|Dentrix G5/G6/G7 & Higher||Direct SDK Bridge for EzDenti- Must update to G7||Full Integration||Bridge|
|Dentrix Ascend (Cloud based)||This program can integrate with Dexis only and then use twain* no support for datagrabber currently availible||NS||Twain||AutoAdvance|
|Dexis||AutoAdvance in Dexis 10 or higher only.||NS||Twain||Twain /AutoAdvance|
|Dexis imaging (Cloud based) (basically is Dentrix Ascend) Line 36||Twain||AutoAdvance|
|Diamond Dental Software (DDS)*||DD, DIA-XV|
|Dolphin (CLoud)||A few sites have inidcated it does not always work stream lined after each scan they have to close Doulphin cloud after maybe 1 or 2 scans.||NS||Twain|
|Dolphin||See legacy SDK chart.||NS||Twain||Twain|
|DSN/Perio-Exec||Bridge avialilbe for Easydent4, *****EzDent-i bridge is Beta for version 16 and higher only.||Bridge||Bridge|
|Eaglesoft *||Full Integration||Twain||AutoAdvance|
|Easy Dental||Full Integration||Bridge|
|EasyDent aka EzDent *||Full Integration|
|Evidentiae||Twain has been verified||Twain||Twain|
|Exact Dental||Full Integration|
|First Pacific Corporation (Aspire)||Aspire is the PMS name||Full Integration, DIA-XV|
|SOFTWARE NAME I – Z||Special Nots:||Apteryx/DG Bridge||EasyDent||EzDent I||NCSW||HD Sensor|
|ICE Classic (Web based IE IMS)||.JPG or .PNG Save format*||NS||Twain||Twain|
|ImageXL (by PEB Software)||Requires Special setup* by PEB, does not pass gender||Bridge||Bridge||Bridge/Twain||SDK/Twain|
|Logic Tech Paradigm||DIA-XV|
|MacPractice||(running windows Parallels)||DIA-XV|
|Medims||Last known version in 2009*||Bridge|
|MiPacs (Hospital Grade Server set) also GEPACS*||Bridge: AnyGateway
Twain: Using Dental Enterprise Viewer*
|MOGO||Has direct bridge available for EzDent-i VTEzBridge32.exe, must set path for it to work by checking "Vatech"box then click cmd to set path||S, DIA-XV|
|Oasys - Crossfire||*Need confirmation||Twain||Twain|
|OMS Vision/DentalVision||Old OMS/DentVision Versions only*||DIA-XV||Bridge|
|OpenDental *||Has free bridge under "preference"||Full Integration||Bridge||Bridge|
|Oral Exe/Dental Exec/Perio Exec||DIA-XV|
|Oral Surgery Exec||Oral Exec-Dental Exec-Perio Exec||S|
|Ortho2||Ortho II is known to not work well with twain; and does not auto import; it will work but with special user sequence to manually import the image post-capture.||TNS*||Twain||Twain|
|OrthoTrack||End User Must have SQL login for connection.||Full Integration||Twain||Twain|
|Patient Base||Full Integration|
|PBS Endo||PBS Endo has their own bridge to XV.||DIA-XV|
|Perfect Byte ( PByte )||Full Integration|
|Planet DDS||Including Denticon Web based||DIA-XV||DIA-XV||Twain||Twain|
|Practice Web||Resold version of OpenDental||Full Integration||Bridge||Bridge|
|Practice Works *||If XV is used we can bridge from XV (With EU updating any missing demographics manually in XV First) - other wise no support*
" - EzDent-i uses only sniffer to pass but must be updated version of EzDent-i.
|Radiology Lab Management Systems (RLMS)||DD|
|SiDexis||Twain does not properly function in SiDexis - Manual import only||Not Supported||Not Supported|
|Suzy Dental4||SDK* patient data into Easydent4*||Bridge|
|The Digital Office (TDO)||DD|
|TIger View||*Most capture programs should twain* - Tiger View Classic is no longer supported; TigerView will charge $2K for support.||Twain||Twain - Single capture only.|
|Tiger View Pro||*Auto Advance is supported per TigerView support||Twain||Twain*|
|Tops Ortho (All versions)||Tops is a MAC based system and is not able to integrate to windows platforms, the only option we have is to export the image to a file directory on the network somewhere that Tops is also able to see and manually import the file* - NO twain support||NS||NS||NS|
|Tracker||Bridge availilbe and setup/supported by Tracker directly*||Bridge/Twain||Bridge/Twain|
|VixWin (all versions)||*Must use the older twain DS type to work, does not support newer twain version||Twain||Twain Single capture only.|
|Visix (Air Techniques)||Visix, which is a rebranded version of Tigerview.||Twain||Twain Single capture only.|
|Windent||Direct integration does not apply to KDI versions||S|
|WinOMS Cloud (Citrix)||Special Setup required||S||Twain||Twain|
|WinOMS||Easydetn4- No support: Same as Practice works "If XV is used we can bridge from XV (With EU updating any missing demographics manually in XV First) - other wise no support for Easydent4*
- EzDent-i : uses only sniffer to pass but must be updated version of EzDent-i. ; Patient will pass data but will not pass Gender but will add the patient , the end user will have to input /modify the chart to add this missing information before scanning patient else an error will occur. .
|XLDent by PEB||Does not pull Gender||S,DIA-XV|
|XNAV (GUIDE Software)||Uses login TSDF to access data*||Bridge|
To use the Integration Chart, first locate the Software in the Software Name Column. 1, Next you will have any special notes such as limited functionality or anything that has been recorded which is important to note. Each column after this represents the corresponding key below and their meaning.
1. Apteryx/DG Bridge: This section indicates what functionality Datagrabber or Apteryx Program such as XrayVision is capable of obtaining from the corresponding program to it self. Typically if using Datagrabber to passing information to Vatech programs all modes in this section with the exception of limited Sniffer will work and bridge patient information in some manner to Vatech Software. Vatech Software is fully compatible with Apteryx program, the limitation is what Apteryx is able to obtain and pass alone.
Bridge= Built in Bridge or Button from Calling Application to Apteryx/XVvision or other rebranded version IMS (No Datagrabber Support).
Full Integration= Full bridge functionality support via Datagrabber (To 3rd party such as Vatech Software) or SDK (To XrayVision/Rebranded)
Datagrabber= DataDrill functionality only to 3rd party software via Datagrabber. This indicates all required fields are obtainable via Datagrabber to pass to Vatech Software.
Sniffer= Sniffer only - indicates limited ability to pull data, typically does not pull all required data, manual imput of missing data required once charting is passed.
Note: Only EzDent-i supports the use of sniffer, Easydent4 does NOT support the use of Sniffer. It can only be used once the chart is manually entered to pull up the chart afterwards.
NS= No support
2. Easydent4/EzDent-i - This column represents the ability for the calling appication to bridge directly passing patient information to add or open charting in Vatech Software.
B= Built in Bridge or Button from Calling Application directly into Easydent4 or EzDent-i.
NS= No support
3. NCSW (SDK) Capture Program / HD Sensor - This column represents the ability of the calling application listed to directly call to the capture program either by SDK - passing patient information and saving directly or using Twain.
Note: Not all programs capable of twaining are listed, those listed are simply one's that have been confirmed to work. Most twain compliant programs will support twain*.
T= Twain 2D images directly into the calling application. No patient information is shared.
B= Bridge directly from calling application, or SDK to Capture Software. .
NS= No support
Note: Please contact those software's that are not listed to confirm they support twain.
All listed are subject to change with out notice, Vatech America does not have any control over the functionality of any linking software SDK or Datagrabber useage. Please review the following information for proper support channels.