- Vatech 3D Viewers
- Hardware Sensors & EzRay Air P
- Hardware Capture Program
- Legacy Devices
- General Q&A
Please use the following search box to quickly locate valuable FAQ support guide.You can use key words or full statement to search by. Alternatively you can simply browse by Category below
Select the Pax-i PCH2500
Download the Pax-i Twain Wrapping Module.
Extract the zip and follow the included instructions.
For new units having a manufacture date in or after 2017
For units having manufacture date in or after 2017 - NCSW comes by default with twain included on the Installation CD. SImply select twain mode as the save type option during installation to install the twain.
The twain files will install into the PC and the default capture settings for twain mode will be automatically generated.
You will only need to open the configuration file to point to proper patient information in order for the capture program not to error.
Browse to C:\VCaptureSW\Settings Files\ and open the 'Environment.ini'
Once open scroll down to the DB Section settings, and locate the [input_settings] section for the chart read path.
Verify the read path is set to as shown above. The default location is C:\VCaptureSW\ , in that case change to C:\VCaptureSW\Exe\ as shown , save and exit.